FlexDAQ Data Logger
Ideal for remote monitoring in both common and demanding geotechnical conditions
Ideal for remote monitoring in both common and demanding geotechnical conditions, RST FlexDAQ Data Loggers offer precise measurement and reliable data acquisition from various sensor types and gauges including vibrating wire, thermistor, MEMS (analog and digital), Tensmeg, linear potentiometer, strain gauge, LVDT, TDR, etc.
The 4 main FlexDAQ models are the CR6, CR300, CR800, and CR1000X. All offer extreme flexibility in their design configurations and are custom-made to accommodate a variety of sensor types as they pertain to the parameters of your project. The framework for building a FlexDAQ Data Logger is dependent on the type, number, precision, and speed of measurements required. Best of all, FlexDAQ Data Loggers arrive to you completely pre-assembled, pre-wired, pre-tested, and pre-programmed; ready to be put to work straight out of the box with minimal set-up.
All FlexDAQ Data Loggers bear similarities in measurement and programming capabilities and can easily incorporate additional sensor and telecommunication options.
|Power||12 Vdc (under load), unregulated|
|Current Drain||10µ quiescent; 8 mA active|
|Reset Active Levels, max.||2.0 V|
|Clock Active Levels, max.||2.0 V|
|Min. Clock Pulse Width||1 ms|
|Max. Actuation Relay Time||20 ms|
|Relay Operation||Break before make|
|Initial Relay Resistance, max.||0.1 Ohm|
|Max. Switching Current||1 A|
|Min. Contact Life||107 closures|
|Operating Temp.||-40°C to 70°C
(-40°F to 158°F) – extended
|Size||8.15”H x 1.05”W x 3.17”D|
|Weight||0.24 kg (0.53 lbs.)|
FlexDAQ Data Loggers are commonly used for:
View illustrations on how FlexDAQ Data Loggers are used for:
NOTE: Due to the customizable nature of each FlexDAQ Data Logger model, specifications can vary greatly in accordance to customer specifications and optional additions. Please contact RST Instruments Ltd. for complete information.
FlexDAQ Data Logger
RST Flexi-Mux Instruction Manual
FlexDAQ Setup and Installation Manual
I’m very happy with the quality of the instrumentation and technical support from RST, as always. They worked with us from the initial stages of the project to ensure that we were fully prepared and equipped to use the RSTAR network for the first time. They delivered everything within the tight project time frame to ensure that there were no delays on site, and the submersible tilt meters looked like new even after over twelve months of being submerged in Thames water!
-Richard Lipscombe, BEng (Hons), Director & Principle Consultant - RL Geotechnical Ltd