Submersible Tilt Meter
The RST Submersible Tiltmeter provides precision real-time remote monitoring of tilt of submerged structures.
The RST Submersible Tiltmeter provides precision real-time remote monitoring of tilt of submerged structures. It consists of a MEMS inclinometer sensor and electronics mounted inside a rugged waterproof enclosure. The instrument housing is machined from solid stainless steel, providing extreme endurance for long-term high-pressure underwater service. The cable entry is a submarine grade connector, which provides watertight performance at depths exceeding 200 meters.
|Resolution (digital)||±2 arc sec. (±0.0006°) (0.01 mm/m)|
|Resolution (analog)||±5 arc sec. (±0.025 mm/m) (10Hz BW)|
|Non-linearity (digital)||±0.0125% F.S. (±0.002°) (0.03 mm/m)|
|Non-linearity (analog)||±0.05% F.S. (±0.0075°) (0.13 mm/m)|
|Repeatability (digital)||±0.0125% F.S. (±0.002°) (0.03 mm/m)|
|Repeatability (analog)||±0.025% F.S. (±0.004°) (0.06 mm/m)|
|Sensor||MEMS (Micro-Electro-Mechanical Systems) Accelerometer, Uniaxial or Biaxial|
|Operating Temp.||-40 to 85°C (-40 to 185°F)|
Submersible Tilt Meters are commonly used for:
|MEMS Submersible Uniaxial Tiltmeter Digital Output||IC6750|
|MEMS Submersible Uniaxial Tiltmeter 4-20mA Output||IC6752|
|MEMS Submersible Biaxial Tiltmeter Digital Output||IC6756|
|MEMS Submersible Biaxial Tiltmeter 4-20mA Output||IC6753|
|Signal Cable: 4 conductor 22 AWG OSD with polyurethane jacket||EL380004|
|Horizontal Mounting Plate – MEMS Submersible Tiltmeter||IC6760|
|Vertical Mounting Bracket – MEMS Submersible Tiltmeter||IC6761|
|flexDAQ Dataloggers (analog and digital systems)|
I’m very happy with the quality of the instrumentation and technical support from RST, as always. They worked with us from the initial stages of the project to ensure that we were fully prepared and equipped to use the RSTAR network for the first time. They delivered everything within the tight project time frame to ensure that there were no delays on site, and the submersible tilt meters looked like new even after over twelve months of being submerged in Thames water!
-Richard Lipscombe, BEng (Hons), Director & Principle Consultant - RL Geotechnical Ltd